- Poster presentation
- Open Access
Is the measurement of delivered dose a good performance measure? A characterization of delivered and respirable delivered dose in two brands of jet nebulizer
Clinical and Translational Allergy volume 3, Article number: P10 (2013)
Use of a nebulizer facilitates rapid administration of high doses of drug, which is desirable for patients presenting with acute severe asthma. Breath-enhanced jet nebulizers direct the patient’ s air flow through the nebulizer handset during inhalation producing aerosol output with a more favorable respirable profile. SideStream Plus (SS+; Philips Respironics) and Nebutech (NB; Salter Labs) are breath-enhanced jet nebulizers, designed to produce a high delivered dose in a short treatment time. A previous in vitro characterization of delivered dose and droplet size distribution of arformoterol nebulized via the SS+ and the NB indicated that the SS+ performed favorably, compared with the NB. We present results of an in vitro characterization of salbutamol sulphate (albuterol sulfate) delivery via the SS+ nebulizer and the NB nebulizer.
Each nebulizer was filled with 3 ml salbutamol sulphate (5 mg/2.5 ml) and run with a driving flow of 8 l/min into a CEN adult tidal breathing pattern (Vt; 500 ml, 15 BPM, I:E ratio; 1:1). The nebulizers were run in triplicate to sputter point and sputter point plus 60 s. Dose delivered to a filter (delivered dose; DD), placed between the nebulizer and breathing emulator, was quantified using high performance liquid chromatography and expressed as a µl solution equivalent. A laser diffractor (Malvern Spraytec) was used to assess fine particle fraction (FPF) and mass median diameter (MMD) after 60 s nebulization time. Fine particle dose (FPD) was calculated (FPF x delivered dose).
The results are shown in Table 1
The total DD from the 2 nebulizers was comparable. Aerosol delivered via the SS+ nebulizer had a lower MMD and higher FPF, compared with that delivered via the NB nebulizer. This resulted in a 74% higher FPD delivered to sputter, and a 55% higher FPD delivered to sputter plus 60 s from the SS+ nebulizer compared with the NB nebulizer. The FPD represents respirable aerosol, and is therefore more clinically relevant than the DD. These results are consistent with previous work, and indicate that a high respirable dose of aerosol is produced by the SS+ nebulizer .
Bauer A: Respir Care. 2009, 54 (10): 1342-1347.
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Rehman, M., Metcalf, A. & Hatley, R. Is the measurement of delivered dose a good performance measure? A characterization of delivered and respirable delivered dose in two brands of jet nebulizer. Clin Transl Allergy 3 (Suppl 1), P10 (2013). https://doi.org/10.1186/2045-7022-3-S1-P10
- Droplet Size Distribution
- Fine Particle Fraction
- Acute Severe Asthma
- Salbutamol Sulphate